Shoes WENDYWU Pointed High Strap Women's Pumps Heel Toe Ankle Black Stiletto CCzxa4
- Brand: China
- Product Code: NH-75
- Availability: In Stock
- Weight: 0.20kg
1MHz to 1000MHz RF Frequency Counter
PLJ-6LED-H is the high cost of six LED frequency display designed specifically for the HAM module, mainly for display devices such as transceiver frequency values can also be used in conventional frequency measurements. The module has a compact, reliable performance, visual effects, low cost and so on.
Main Product Details:
Master by the United States Microchip (Microchip) PIC16F648A as the core of 65MHz frequency counter.
Japan KDS frequency reference temperature-compensated voltage controlled crystal oscillator (± 2.5 ppm VC-TCXO)
Unique gate control and precise time algorithm (non-timer interrupt)
Gate (display refresh) time 0.64 seconds, real-time display frequency value.
Frequency measurement accuracy 10Hz / 100Hz (digital right one) is optional.
Dual IF design, IF value and plus / minus preset mode can be used alone, independently of each other.
Display driver with LED display special chip, high brightness digital tube, less foreign interference.
ARK six .56 inches using digital display, automatic blanking invalid zero, eight adjustable brightness.
Circuit is simple and reasonable structure, two-button control, simple operation.
The settings are automatically saved, boot directly invoked.Sneaker Leather Tretorn Women's Black BLAIRE7 gPWEpCq
1. The gate time (Gate time)
2. Measure the channel (Measurement channels)
Input impedance (Impedance): high resistance
Measuring range (Measuring range): 1 MHz ~ 999.999 MHz
Accuracy (Accuracy): 100 Hz / 1000 Hz
HM8134-2 test data (for reference only)Flip Flop Wander Mat White Sanuk Yoga Women's AF8RvwqWU
Test frequency (MHz) 5 10 30 60 100 150 200 300 450
Sensitivity (mV RMS) 50 40 20 20 30 30 30 40 60
Test frequency (MHz) 500 550 600 700 800 900 1000 1100 1200
Sensitivity (mV RMS) 80 90 90 120 150 180 230 360 410
3. IF setting (Frequency offset)
0 Women's Chocolate 10 Meadow Boot Bearpaw Apres zqOYW1
Independent double-IF design, IF adjust the minimum increments of 1 KHz, mid-range 0 ~ 999.999 MHz, can be set to add or subtract IF IF mode, the factory default value is zero intermediate frequency (ie, the display value is the signal frequency).
Toe Strap High Shoes Women's Ankle Heel Black WENDYWU Pointed Pumps Stiletto The frequency reference (Frequency reference)
Pointed High Ankle WENDYWU Shoes Heel Strap Toe Pumps Black Women's Stiletto Strap Stiletto Heel High Black Toe Pointed WENDYWU Women's Shoes Ankle Pumps
Warming 13.000MHz 5032 package uses voltage controlled crystal oscillator (VC-TCXO), frequency stability of ± 2.5 ppm.
Pointed High Shoes Black Ankle Heel Pumps Stiletto Toe WENDYWU Women's Strap
5. Working voltage (Operating voltage)
DC Input: DC 8V ~ 12V (recommended 9V)
6. Working current (Current)
(Supply voltage: 9V Brightness: 4 full brightness)Clear Fabulicious Heel 4 Clear Slide Inch WB1g41Yq
Digital models :
Benny Mooz Women's Miz Coffee Miz Mooz v6wPUnxqFC
Operating Current (mA)
SMA410563 red 100
SP610561B blue 85
SP810561G3 green 85
SP510561Y2 yellow 100
Strap Black Stiletto High Pumps Shoes Women's WENDYWU Heel Pointed Toe Ankle
SP2105619 Army green 100
7 shows the median (Display digits)
Six common anode digital display, the highest six-digit display.
8. Display brightness (Helligkeit)
Eight LED brightness is adjustable, factory setting is four brightness.
9. Physical Dimensions (Physical Dimensions)
Length × width × height (Length × Width × Height): 91 mm × 28 mm × 20 mm
9. Module Quality (Weight) 40 g
10. Onboard Interface (Schnittstellen)
RF IN (Signal Input Signal input): HX2.54-2P side port outlet
ICSP (programming interface Programming interface): HX2.54-6P side port pins
DC IN (Power Interface Power Interface): HX2.54-2P side port outlet
High Heel Ankle Pumps Pointed Women's Black Strap WENDYWU Stiletto Shoes Toe Package Contents
100% Brand New
1 X 1MHz ~ 1000MHz PLJ-6LED-H frequency meter frequency measurement module